Measures the size and radius of HSK tool holder drive keys
Available for HSK 32, 40, 50, 63, 80, 100, 125, 160
Complete set with case includes measuring ring with dial indicator, measuring gauge for lower deviation, measuring gauge for upper deviation, mean deviation and setting gauge for dial indicator.
Function The setting gauge is placed into the gauge ring, and the dial indicator is set to zero. The measuring gauge 1 (second gauge pictured) is then used to measure the lower deviation of the drive key, and the measuring gauge 2 (third gauge) for upper deviation.
For more information regarding this item (Tool Holder Drive Key Tool Taper Gauges) or other items, fill out the form below
or contact our office directly:
Location: 304 North Main St, Suite 104, Rockford, IL 61101-1101 USA
Measuring gauge for HSK tool holder taper drive key slots according to ISO 12164-1 (comparison measurement). Download or print PDF data sheet for complete instructions.
When inserted in the spindle, the "go" gauge should be in contact with the plane face. The "no go" gauge should not contact the face. Two measuring gauges in storage case: "go" and "no go" plug gauges.
TAC Rockford Morse tool holder taper gauges are for inspection of Morse tool holder tapers. Gauges made from a hardened gauge steel, and include a padded carrying case and certificate. Manufactured per DIN 229.
Both measuring gauges are set to zero with the calibrated taper plug gauge. When an HSK tool holder is inserted in the ring gauges, deviations in diameters d2 or d3 are shown on the indicator.
Gauging kit for HSK tool holder taper with pneumatic measuring column. The Multifunctional Measuring and Control Device is designed for high precision and multifunctional capabilities. In addition to providing various ways for measuring input, it can be customized to meet customer requirements.
Series 968 gauges offer comprehensive dimensional verification of HSK tool holders.
Available for Steep Taper 30, 40, 50, and 60. Includes padded storage case.
Using the calibrated master, the distance of the 30 degree clamping slope relative to l 2 is preset using the dial indicator holder. The tool taper to be measured is placed in the centering piece, allowing the variation of the taper to be measured.